The EMERGENCY ALARM PANEL monitors the network pressure of medical gases and vacuum:
nitrus oxide N2O
carbon dioxide CO2
compressed air Air
The membrane keyboard consists of an embossed aluminium frame, coulour RAL 9002 with powder coating and a electronic unit fixed with the frame via fixing strips.
The easy and inexpensive Mounting is allowed by a patented locking technology in connection with the mounting frame on the device mounting housings.
The membrane keyboard can show 1-3 emergency signals for series 3EN and 1-5 emergency signals for series 5EN.
Marking the emergency signals by pushing the labels into the labeling fields being behind the foil for hygenic reasons.
The patented caption allows for easy replacement.
In addition, the emergency signal devices can be used for all industrial gas supply systems.
The 3-digit seven-segment LED displays show the actual pressure (in “bar”) measured by pressure transducers in the monitored area. Additionally a 5 LED trend display, signalises compliance to the normal line pressure or a deviation from it.
One or more Repeater Alarm Panels of the CMN and CMNG series can be connected using a two-wire line (respectively a four-wire when also supplying power to the Repeater Alarm).
Optionally, the Emergency Alarm Panels can be equipped with an internal interface (e.g. modbus or RS485) that allows the forwarding of all alarms to a bus system.
The panels will be pre-configured at the factory. Using a programming device (CMS/PG7576) the pressure values for nominal pressure, pre warning and alarm can be reconfigured.
EN-Devices - ENG-Devices
EN - Devices
5EN - Flush Mounting
monitoring of up to 5 gases
6EN - Flush Mounting
monitoring of up to 6 gases
Reset & Test
Für mehr Informationen rufen Sie bitte die Desktop-Version am PC auf.
5EN/6EN - Flush Mounting
5ENG/6ENG - On-Wall Enclosure
depth (incl. Alarm Panel):
depth with built-in transformer:
Flush Mounting - Hollow-Wall Mounting - Control Panel Mounting
Use in Area Shut-Off Valve Unit
Monitoring of pressure values by pressure Transmitter and alarming at critical pressure value.
You can discover all the information of our products:
- tender texts
- instructions for use
- wiring diagrams
- instructions for mounting
- failure analysis / malfunction cause / fault isolation / repair
- instructions for adjustment
- acceptance reports
- significant maintenance steps
- protocol according to EN standard
- maintenance interval
- special equipment